Citation, DOI, disclosures and article data
At the time the article was created Ayush Goel had no recorded disclosures.View Ayush Goel's current disclosures
At the time the article was last revised Craig Hacking had the following disclosures:
- Philips Australia, Paid speaker at Philips Spectral CT events (ongoing)
These were assessed during peer review and were determined to not be relevant to the changes that were made.View Craig Hacking's current disclosures
Doppler shift or Doppler effect is defined as the change in frequency of sound wave due to a reflector moving towards or away from an object, which in the case of ultrasound is the transducer.
When sound of a given frequency is discharged and subsequently reflected from a source that is not in motion, the frequency of the returning sound waves will equal the frequency at which they were emitted.
However, if the reflecting source is in motion either toward or away from the emitting source (e.g. an ultrasound transducer) the frequency of the sound waves received will be higher (positive Doppler shift) or lower (negative Doppler shift) than the frequency at which they were emitted, respectively 2.
positive Doppler shift
negative Doppler shift
frequency of received sound waves < frequency of emitted sound waves
source reflecting sound waves is moving away from the emitting source
depicted in color flow Doppler as blue
spectral envelope (in continuous and pulsed wave Doppler) below the baseline
F = 2fo(v/c)cos(Q)
F is Doppler frequency shift
fo is transmitted frequency from ultrasound probe
v is the velocity of moving reflector
c is the velocity of sound in the medium
Q is the angle between ultrasound beam and axis of flow
the Greek letter theta (θ) is also used
The above doppler formula is used because the transducer is not parallel to the axis of the moving object 4.
The magnitude of the Doppler shift is affected by the angle at which the reflecting source is traveling in relation to the transmitting source. This is accounted for in the Doppler equation with the "cosine(θ)" parameter; the maximum Doppler shift occurs when the relative motion occurs at a Doppler angle of 0 degrees (the cosine of 0 = 1) and no Doppler shift will be noted when the motion of the reflecting source is perpendicular (cosine of 90 = 0) 3.
History and etymology
Named after Austrian physicist, Christian Andreas Doppler (1803-1853) 1.
- 1. Katsi V, Felekos I, Kallikazaros I. Christian Andreas Doppler: A legendary man inspired by the dazzling light of the stars. (2013) Hippokratia. 17 (2): 113-4. Pubmed
- 2. Kruskal JB, Newman PA, Sammons LG, Kane RA. Optimizing Doppler and color flow US: application to hepatic sonography. (2004) Radiographics : a review publication of the Radiological Society of North America, Inc. 24 (3): 657-75. doi:10.1148/rg.243035139 - Pubmed
- 3. Ravi Rasalingam, Majesh Makan, Julio E. Perez. The Washington Manual of Echocardiography. (2012) ISBN: 9781451113402
- 4. Keane P, Ruiz-Garcia H, Sadda S. Advanced Imaging Technologies. Retina. 2013;:133-50. doi:10.1016/b978-1-4557-0737-9.00005-9