Dose length product
Citation, DOI, disclosures and article data
At the time the article was created Matt A. Morgan had no recorded disclosures.View Matt A. Morgan's current disclosures
Dose length product (DLP) measured in mGy*cm is a measure of CT tube radiation output/exposure. It is related to volume CT dose index (CTDIvol), but CTDIvol represents the dose through a slice of an appropriate phantom. DLP accounts for the length of radiation output along the z-axis (the long axis of the patient).
DLP = (CTDIvol ) * (length of scan, cm)
DLP does not take the size of the patient into account and is not a measure of absorbed dose. If the AP and lateral dimensions of the patient are available, then the size specific dose estimate (SSDE) can be used to estimate the absorbed dose.
It is important to remember that the dose length product is not the patient's effective dose. The effective dose depends on other factors including patient size and the region of the body being scanned. Some multipliers, called k-factors, have been estimated to convert DLPs into effective doses, depending on the body region. If interested, consult reference 3.
- 1. Jerrold T. Bushberg, John M. Boone. The Essential Physics of Medical Imaging. ISBN: 9780781780575
- 2. Durand DJ, Mahesh M. Understanding CT dose display. (2012) Journal of the American College of Radiology : JACR. 9 (9): 669-71. doi:10.1016/j.jacr.2012.06.003 - Pubmed
- 3. Walter Huda, Kent M. Ogden, Mohammad R. Khorasani. Converting Dose-Length Product to Effective Dose at CT. (2008) Radiology. 248 (3): 995-1003. doi:10.1148/radiol.2483071964 - Pubmed