Dual-energy CT (DECT) utilizes two separate energy sets to examine the different attenuation properties of matter, having a significant advantage over traditional single energy CT. Independent attenuation values at two energy sets can create virtual non-contrast images from contrast-enhanced imaging as well as delineate the composition of renal calculi and arterial plaque 13.
There are different DECT technologies available:
- two x-ray tubes producing different voltages offset at approximately 90°
- single-source helical
- two spiral scans are consecutively acquired at different tube potentials.
- single-source twin-beam
- two-material filter splits the x-ray beam into high-energy and low-energy spectra on the z-axis before it reaches the patient.
- single-source sequential
- each x-ray tube rotation is performed at high- and low- tube potential.
- single-source rapid switching
- the x-ray tube switches between high- and low- tube potential multiple times within the same rotation.
- dual-layer DECT
- a "sandwich" detector absorbs high-energy photons with its top layer, and low-energy photons with its bottom layer
When an atom undergoes the photoelectric effect, the electron from that respected K-shell otherwise referred to as the inner shell is ejected via the incident photon. As that electron is excited, vacant space is 'filled' by a neighboring electron, releasing energy as a photoelectron.
In short, when a photon has sufficient energy to overcome the electron's binding energy in the K-shell, that atom undergoes the photoelectric effect.
Each substance owns a unique K-shell binding energy; known as the K-edge. There is a significant spike in attenuation that results just beyond the energy of the K-edge, this peak is unique to every material and holds valuable information about the substance's composition.
The different photoelectric energies and K-edges are the bread and butter of dual-energy CT. Although most elements in the human body have very low K-edges (0.01-0.53 keV), elements like iodine and calcium have K-edges of 33.2 keV and 4.0 keV respectively, making them sufficiently larger than surrounding structures and are particularly important in the clinical setting 1-3.
For instance, at 80 kVp a structure that contains no (introduced) iodine, such as the liver, has an attenuation based on its K-edge of x, yet when iodine (33.2 keV) is introduced into that same structure, it has a higher attenuation of y bringing it closer to 80 kVp.
As 80 kVp is closer to 33.2 keV than 140 kVp, the structures containing iodine will retain less attenuation as the kVp progressed beyond the K-edge of iodine. Therefore, when using two energies, it is possible to delineate structures based solely on their attenuation differences between 80 kVp and 140 kVp.
A dual x-ray source, tube A (140 kVp) and tube B (80 kVp or 100 kVp) with an angular offset of 90 degrees are preferred offsets for a dual source scanner in the current literature 1-5.
- 1. Vlahos I, Godoy MC, Naidich DP. Dual-energy computed tomography imaging of the aorta. J Thorac Imaging. 2010;25 (4): 289-300. doi:10.1097/RTI.0b013e3181dc2b4c - Pubmed citation
- 2. Godoy MC, Heller SL, Naidich DP et-al. Dual-energy MDCT: comparison of pulmonary artery enhancement on dedicated CT pulmonary angiography, routine and low contrast volume studies. Eur J Radiol. 2011;79 (2): e11-7. doi:10.1016/j.ejrad.2009.12.030 - Pubmed citation
- 3. Lu GM, Wu SY, Yeh BM et-al. Dual-energy computed tomography in pulmonary embolism. Br J Radiol. 2010;83 (992): 707-18. doi:10.1259/bjr/16337436 - Free text at pubmed - Pubmed citation
- 4. Ascenti G, Mazziotti S, Lamberto S, Bottari A, Caloggero S, Racchiusa S, Mileto A, Scribano E. Dual-energy CT for detection of endoleaks after endovascular abdominal aneurysm repair: usefulness of colored iodine overlay. Am J Roentgenol. 2011 Jun;196(6):1408-14. doi: 10.2214/AJR.10.4505.
- 5. Johnson TR. Dual-energy CT: general principles. AJR Am J Roentgenol. 2012;199 (5_supplement): S3-8. doi:10.2214/AJR.12.9116 - Pubmed citation
- 6. Rutherford RA, Pullan BR, Isherwood I. Measurement of effective atomic number and electron density using an EMI scanner. Neuroradiology. 1976;11 (1): 15-21. Pubmed citation
- 7. Coursey CA, Nelson RC, Boll DT et-al. Dual-energy multidetector CT: how does it work, what can it tell us, and when can we use it in abdominopelvic imaging?. Radiographics. 2010;30 (4): 1037-55. doi:10.1148/rg.304095175 - Pubmed citation
- 8. Reagan AC, Mallinson PI, O'Connell T et-al. Dual-energy computed tomographic virtual noncalcium algorithm for detection of bone marrow edema in acute fractures: early experiences. J Comput Assist Tomogr. 2014;38 (5): 802-5. doi:10.1097/RCT.0000000000000107 - Pubmed citation
- 9. Pache G, Krauss B, Strohm P et-al. Dual-energy CT virtual noncalcium technique: detecting posttraumatic bone marrow lesions-feasibility study. Radiology. 2010;256 (2): 617-24. doi:10.1148/radiol.10091230 - Pubmed citation
- 10. Guggenberger R, Gnannt R, Hodler J et-al. Diagnostic performance of dual-energy CT for the detection of traumatic bone marrow lesions in the ankle: comparison with MR imaging. Radiology. 2012;264 (1): 164-73. doi:10.1148/radiol.12112217 - Pubmed citation
- 11. Stolzmann P, Scheffel H, Rentsch K et-al. Dual-energy computed tomography for the differentiation of uric acid stones: ex vivo performance evaluation. Urol. Res. 2008;36 (3-4): 133-8. doi:10.1007/s00240-008-0140-x - Pubmed citation
- 12. Hidas G, Eliahou R, Duvdevani M et-al. Determination of renal stone composition with dual-energy CT: in vivo analysis and comparison with x-ray diffraction. Radiology. 2010;257 (2): 394-401. doi:10.1148/radiol.10100249 - Pubmed citation
- 13.Machida H, Tanaka I, Fukui R et-al. Dual-Energy Spectral CT: Various Clinical Vascular Applications. Radiographics. 2016;36 (4): 1215-32. doi:10.1148/rg.2016150185 - Pubmed citation
- 14. Otrakji A, Digumarthy SR, Lo Gullo R, Flores EJ, Shepard JA, Kalra MK. Dual-Energy CT: Spectrum of Thoracic Abnormalities. Radiographics : a review publication of the Radiological Society of North America, Inc. 36 (1): 38-52. doi:10.1148/rg.2016150081 - Pubmed
- 15. Lourenco PDM, Rawski R, Mohammed MF, Khosa F, Nicolaou S, McLaughlin P. Dual-Energy CT Iodine Mapping and 40-keV Monoenergetic Applications in the Diagnosis of Acute Bowel Ischemia. (2018) AJR. American journal of roentgenology. 211 (3): 564-570. doi:10.2214/AJR.18.19554 - Pubmed
Related Radiopaedia articles
Physics and imaging technology: CT
computed tomography (CT)
- CT technology
- dual energy CT
- CT image reconstruction
- CT image quality
- CT dose
- CT contrast medium
- coronary CT angiography
- patient-based artifacts
- physics-based artifacts
- hardware-based artifacts
- CT safety
- history of CT