Dynamic susceptibility contrast (DSC) MR perfusion
Citation, DOI & article data
Dynamic susceptibility contrast (DSC) MR perfusion is one of the most frequently used techniques for MRI perfusion, and relies on the susceptibility induced signal loss on T2*-weighted sequences which results from a bolus of gadolinium-based contrast passing through a capillary bed. The most commonly calculated parameters are rCBV, rCBF, and MTT.
This technique is sometimes referred to, perhaps more accurately, as dynamic susceptibility contrast-enhanced MR perfusion, still abbreviated to DSC. This should not be confused with dynamic contrast-enhanced (DCE) MR perfusion, which relies on T1 shortening due to gadolinium-based contrast.
Physics and technique
DSC perfusion exploits the regional susceptibility-induced signal loss caused by paramagnetic contrast agents (such as commonly used gadolinium-based compounds) on T2-weighted images 1,2. Although this technique can be performed with both T2 (e.g. spin echo) and T2* (e.g. gradient-echo echo-planar) sequences, the former requires higher doses of contrast, which is why T2* techniques are more commonly employed 2.
A bolus of gadolinium-containing contrast is injected intravenously and rapid repeated imaging of the tissue (most commonly brain) is performed during the first pass. This leads to a series of images with the signal in each voxel representing intrinsic tissue T2/T2* signal attenuated by susceptibility-induced signal loss proportional to the amount of contrast primarily in the microvasculature 1,2.
After image acquisition, a region's signal is interrogated over the time-course of the perfusion sequence, generating a signal intensity-time curve, from which various parameters can be calculated (e.g. rCBV, rCBF, MTT). These values can then be used to create color maps of regional perfusion.
Because this technique relies upon detecting signal loss due to small amounts of contrast, if there is significant signal loss due to the presence of calcification or blood products, or due to artifact from adjacent dense bone or aerated sinuses, obtained values will not be reliable. Similarly, values in a region immediately adjacent to large vessels will also be affected.
- 1. Essig M, Shiroishi MS, Nguyen TB et-al. Perfusion MRI: the five most frequently asked technical questions. AJR Am J Roentgenol. 2013;200 (1): 24-34. doi:10.2214/AJR.12.9543 - Free text at pubmed - Pubmed citation
- 2. Petrella JR, Provenzale JM. MR perfusion imaging of the brain: techniques and applications. AJR Am J Roentgenol. 2000;175 (1): 207-19. doi:10.2214/ajr.175.1.1750207 - Pubmed citation
- 3. Clinical MR Neuroimaging. Cambridge University Press. (2009) ISBN:0521515637. Read it at Google Books - Find it at Amazon