High voltage generator
Citation, DOI, disclosures and article data
At the time the article was created Andrew Murphy had no recorded disclosures.View Andrew Murphy's current disclosures
At the time the article was last revised Jeremy Jones had no recorded disclosures.View Jeremy Jones's current disclosures
X-ray units require a high voltage generator to achieve the necessary power required of an x-ray tube. AC power will supply x-ray units with sinusoidal currents, resulting in 'peaks and troughs', limiting an x-ray tube to produce x-rays only half of the 1/60th of s second cycle.
A single-phase high voltage generator converts this AC power into a half or full wave rectified supply with a measure in the thousands of volts.
The half wave rectification results in a peak voltage that will dip to zero, reoccurring; this will consequently have an effect on the behavior of radiation produced and hence the name kilovoltage peak (kVp) was born.
The advancement of high voltage generators from single-phase to three-phase to constant potential generators have overcome this 'voltage ripple' creating a continuous, uninterrupted voltage. Modern x-ray units, which largely utilize, constant potential generators do not have 'voltage ripple' and consequently employ the term kV rather than kVp.
- 1. Stewart C. Bushong. Radiologic Science for Technologists. ISBN: 9780323081351</