Reverse total shoulder arthroplasty (RTSA) is a type of total shoulder replacement. It is often the preferred method when there has been advanced damage to the rotator cuff as seen in rotator cuff arthropathy.
The reverse total shoulder flips the normal mechanical arrangement of the shoulder's ball-and-socket joint. Instead of the humeral head fitting into the glenoid, a rounded hemispheric glenoid prosthesis articulates with a socket on the humeral head prosthesis.
Because of the different mechanical arrangement of the shoulder joint, the deltoid muscle provides most of the movement for the upper extremity after the surgery. Rotator cuff muscles are not as necessary to move the upper extremity, which is preferable in patients with advanced rotator cuff damage. Internal and external rotation are more limited in this arrangement, however.
The indications for reverse total shoulder replacement are expanding and include replacement after tumour surgery and after complex fractures in elderly patients.
The design is thought to have relatively good short and medium-term outcomes compared with other types of total shoulder replacements, with improved recovery time and pain scores 3. Long-term outcomes are not yet well-defined.
- neutral AP view (Grashey view) and axillary views obtained
- >2 mm of lucency around the prosthetic component is concerning for loosening
- "scapular notching" (erosion of the inferior scapular neck) is suggestive of glenoid component failure
- periprosthetic fractures at the base of the acromion at the scapular spine are more common
- better for evaluation if there is a suboptimal radiograph
- as on conventionsl radigraphy, >2 mm of lucency around the prosthetic component is concerning for loosening
- metal reduction techniques have improved the ability to image prostheses on MRI
- useful for imaging the rotator cuff musculature, evaluating the prosthesis dynamically, and evaluating for joint effusion
- cannot be used to evaluate for component loosening
The overall complication rate for all forms of total shoulder arthroplasty is ~15%. Complications include:
- aseptic loosening of the glenoid component: most common complication (~32% of all shoulder replacements, but decreased with reverse total shoulder replacements 5)
- aseptic loosening of the humeral component
- periprosthetic fractures
- hardware failure
- axillary nerve palsy
- infection (uncommon)
- 1. Sanchez-Sotelo J. Total shoulder arthroplasty. Open Orthop J. 2011;5 (1): 106-14. doi:10.2174/1874325001105010106 - Free text at pubmed - Pubmed citation
- 2. Sanchez-Sotelo J. Reverse total shoulder arthroplasty. Clin Anat. 2009;22 (2): 172-82. doi:10.1002/ca.20736 - Pubmed citation
- 3. Frankle M, Siegal S, Pupello D et-al. The Reverse Shoulder Prosthesis for glenohumeral arthritis associated with severe rotator cuff deficiency. A minimum two-year follow-up study of sixty patients. J Bone Joint Surg Am. 2005;87 (8): 1697-705. doi:10.2106/JBJS.D.02813 - Pubmed citation
- 4. Konin GP. Imaging of shoulder arthroplasty. Semin Musculoskelet Radiol. 2015;19 (01): 49-59. doi:10.1055/s-0034-1396767 - Pubmed citation
- 5. Gerber C, Pennington SD, Nyffeler RW. Reverse total shoulder arthroplasty. J Am Acad Orthop Surg. 2009;17 (5): 284-95. Pubmed citation